The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||7 December 2014|
|PDF File Size:||13.37 Mb|
|ePub File Size:||15.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
Keep Away from Live Circuits – Operators must not remove instrument covers. Click Open to start the calibration measurement.
Select the destination folder and input a file name, then click Next button. Replacement Procedure Step 1. Select a test port and corresponding S parameter on which 1-port calibration will be performed. Click OK to proceed. The volume label for C: Touch the x mark on the upper left with your finger.
Atec Agilent-EC User Manual | 60 pages
Turn ON the EC. The manual part number changes when extensive technical changes are incorporated. Figure Recover User Backup ecse Step 5.
Network setting GPIB setting Printer setting The driver for the supported printer installed after purchase is deleted. A chassis terminal; a connection to the instrument’s chassis, which includes all exposed metal structure.
CN or above Step 1. The EC restart automatically. Out-position mmanual push-button switch. The EC restarts automatically.
Atec Agilent-E5071C User Manual
CN or below Returns the contents of the C and D drives to a user-specified state. A message dialog box appears after the completion of the Recovery Image Setup procedure.
Open the screw in clockwise direction. A total of twelve error terms, six each in manuwl forward direction and the reverse direction, are used in a calibration. The procedure consists of the following items. NOTE You need a keyboard for this operation. Select the saved backup image file, then click Next button.
No part of this document may be photocopied, reproduced, or translated to another language without the prior written consent of the Agilent Technologies. DO NOT Service or Adjust the Instrument Alone – Do not attempt internal service or adjustment unless another person, capable of rendering first aid and resuscitation, is present.
The Device Manager e5071 box appears. E Removal Procedure Step 1. Press the standby switch of the EC to turn it on. The System Properties dialog box appears. Press 1, and then press Enter. Inset the key in the lock and turn crockwise. Press any key to continue. Therefore you should follow the procedure which you see amnual the EC.
Click Done to terminate the 1-port calibration process. Click Done to terminate the full 2-port calibration process. In addition to the Windows operating system, the following settings of the EC are returned to the factory state. Touch the x marks in that order with your finger. Recovery function is performed.
A confirmation dialog box is displayed. Also, never save the contents of the drive to the C, D or F drive. It’s a community-based project which helps to repair anything.
Press the button under the screw. After several seconds, the next screen appears automatically even if you do not press any key, so do not miss it. Push the enter to select Yes to start the recovery of the factory backup.
Click Short to start the calibration measurement. Turn the screw in anti-clockwise direction until it is locked. Select the C drive and D drive. Use extreme caution when handling, testing, and adjusting this instrument.
Help File – User Manual for E5071C ENA Network Analyzer (Operation and Programming)
Nanual a while, the EC restart automatically. Select the Manaul drive to restore, then click Next button. A dialog box appears to allow you to select a program to be loaded. Minor corrections and updates that are incorporated at reprint do not cause the date to change. If you perform the following procedure incorrectly, a message asking you whether to return to the previous registration screen and perform the registration appears.
Insert the hard disk into the slot.